This is an archive of papers published by the staff and faculty of Fox Chase Cancer Center. For questions about content, please contact Talbot Research Library
Last updated on
SU-E-T-94: An Advanced Rotating Gamma Ray System for Stereotactic Radiosurgery
Med Phys. 2015 Jun;42(6) :3352
AbstractPURPOSE: Co-60 beams have unique dosimetric properties that are ideally suited for cranial treatments. Co-60 sources with cone-shaped collimators provide conformal dose distributions allowing for ablative treatments with rapid dose falloff to spare nearby critical structures. This work investigates a novel, image-guided, rotational Gamma ray system that provides both superior dose conformity/gradient and accurate stereotaxy for stereotactic radiosurgery (SRS). METHODS: The SupeRay system (Cyber Medical Corp., China) consists of a rotating source chamber containing 30 gamma sources focusing at the isocenter with 4 collimators measuring 3, 4, 8 and 16mm in diameter. A novel switch design enables the 30 Gamma sources to be turned off at any arbitrarily selected 60 degrees interval in order to avoid critical structures. The 3D treatment couch provides automatic treatment positioning between individual shots and the kV imaging system provides orthogonal images with a spatial resolution of 0.24mm to facilitate target localization. Monte Carlo simulations were used to compute dose distributions and compare with measurements and other Gamma ray SRS systems. RESULTS: Monte Carlo results confirmed the SupeRay design parameters including output factors and 3D dose distributions. Its beam penumbra/dose gradient is similar to or slightly better than that of the Elekta Gamma Knife. The penumbra in the (x,y,z) direction was (7.38mm,7.38mm,3.86mm) for the 16mm collimator, (4.83mm,4.83mm,3.12mm) for the 8mm collimator, and (3.03mm,3.03mm,2.38mm) for the 4mm collimator, respectively, on the SupeRay system while it was (9.5mm,10.0mm,2.9mm), (4.3mm,4.3mm,2.9mm) and (3.2mm,3.2mm,1.9mm) for the same collimator sizes, respectively, on the Perfexion system. The kV imaging system together with a non-invasive relocatable frame provides accurate target localization (<0.5mm) for cases requiring multiple treatment fractions. CONCLUSION: Because of the unique dosimetric properties of Co-60 sources and its accurate stereotaxy/dose delivery, SupeRay is ideally suited for single and multiple-session ablative treatments of cranial lesions. This work was partially supported by Cyber Medical Corp.
NotesMa, C Chibani, O Mora, G Li, J Chen, L United States Med Phys. 2015 Jun;42(6):3352. doi: 10.1118/1.4924455.